All Rights Reserved.
EMAT or Electro Magnetic Acoustic Transducer is an Ultrasonic Testing (UT) technique that generates the sound in the part inspected instead of the transducer.
An EMAT induces ultrasonic waves into a test object with two interacting magnetic fields. A relatively high frequency (RF) field generated by electrical coils interacts with a low frequency or static field generated by magnets to generate a Lorentz force in a manner similar to an electric motor.
This disturbance is transferred to the lattice of the material, producing an elastic wave. In a reciprocal process, the interaction of elastic waves in the presence of a magnetic field induces currents in the receiving EMAT coil circuit.
For ferromagnetic conductors, magnetostriction produces additional stresses that enhance the signals to much higher levels than could be obtained by the Lorentz force alone. Various types of waves can be generated using different combinations of RF coils and magnets.
Because the sound is generated in the part inspected instead of the transducer, EMATs have the following advantages over more conventional piezoelectric transducers:
EMAT is capable of generating all wave modes used in ultrasonic testing, including some modes that are very difficult or impractical with conventional piezoelectric transducers. The table below provides a summary guide of the type of wave and technique available for different applications.
As an ultrasonic technique EMAT can be used to meet ISO, AWS, API, MIL-STD and other international standards. Innerspec Technologies has already deployed systems designed to meet the following:
EMAT is also specifically referenced in the following ASTM guides:
Go to top.